← HP3455A teardown - part 1 - calibration module and the backside
A happy working meter
The meter front panel, fully lit up
Your email address will not be published. Required fields are marked *
Save my name, email, and website in this browser for the next time I comment.
This site uses Akismet to reduce spam. Learn how your comment data is processed.