← HP3455A teardown – part 1 – calibration module and the backside
The multimeter backside. Nothing special, but note the filter on the fan – no dust inside at all, it does the job apparently.
Your email address will not be published. Required fields are marked *
Save my name, email, and website in this browser for the next time I comment.
This site uses Akismet to reduce spam. Learn how your comment data is processed.