← HP3455A teardown - part 1 - calibration module and the backside
More snaps of the backside and the cal module cage
More snaps of the backside and the cal module cage... and the obscure interface
Your email address will not be published. Required fields are marked *
Save my name, email, and website in this browser for the next time I comment.
This site uses Akismet to reduce spam. Learn how your comment data is processed.