← HP3455A teardown – part 1 – calibration module and the backside
Another snap of the terminals and the cal module fixture
Another snapshot of the terminals and the calibration module fixture
Your email address will not be published. Required fields are marked *
This site uses Akismet to reduce spam. Learn how your comment data is processed.