← HP3455A teardown – part 1 – calibration module and the backside
Calibration module connector
Calibration module connector – note the special connector in the middle – it’s for the seriously high impedance signals
Your email address will not be published. Required fields are marked *
This site uses Akismet to reduce spam. Learn how your comment data is processed.